Non-Magnetic MRI Dual Pod Ceiling Mount Light
Double Your Effort (x2) Pods: Each Pod: 29" extension arm, 720° rotation & 69" reach to beam center, 359° lamp head/balance arm rotation in horizontal and vertical planes, Unlimited beam placement and Balanced arm rotation. Light has a 30 foot cord to transformer box (see warning below) 9.5 foot cord from transformer to wall outlet.
Certified as MRI Compatible, these surgery lights are ideal for surgical and specialty procedures in MRI facilities. Made of MRI compatible materials and approved by both GE and Siemens for use with their MRI solutions.
Ideal for MRI surgical applications and procedures where maneuverability and compact design are key requirements. Brings performance of a surgical light into the MRI suite. Excellent flexibility and reach combined with high performance and space saving design makes it ideal for the MRI environment. Unlimited positioning and market leading vertical and horizontal reach allow for optimal functionality around the bore. "No Tool" service minimizes risk, and the compact design limits obstruction of the work are around the MRI equipment.
MRI Features (x 2)
Tested and Certified for MRI compatibility
114-inch reach provides head to toe coverage
Drift free balance arm design
(3) 35W long life halogen bulbs
Provides unparalleled reach, range of motion, and beaming flexibility
High intensity medical lighting; illuminates up to 6000 footcandles
Provides head-to-toe coverage
Easy installation and maintenance
Technical Data (x2)
Power Supply: 120V 60Hz AC or 230-240V 50 HZ AC
Light Intensity: Up to 6,000 footcandles/64,500 Lux
Light Source: Halogen Bulb - Dichroic 12V/3x35W/10 degrees
Light Color Temperature: Up to 4,100 kelvin
Medical Standards: FDA Class Class II
Color: White - RAL 9002
Weight: 52 lbs
Warning: The installation guidelines must be strictly followed. The electrical box contains ferrous materials and must NEVER be placed close to the magnet. If the electrical box is moved closer to the magnet at 100 gauss line, electromagnetic interference may be seen, to avoid this
NEVER move the electrical box close to the magnet.
NEVER operate the light while the electrical box is unsecured. The lamp may slightly vibrate.